FACSComp software is used for setting up the FACSCalibur flow cytometer daily. BD CellQuest software is required for analyzing samples and formatting data for subsequent analysis using the FCAP Array software. Setup for the FACSArray Bioanalyzer required only once a month. Sample acquisition is automated on the FACSCalibur, using the carousel loader, whereas the FACSArray instrument acquires the samples directly from the microplate wells. The data are analyzed using the FCAP Array Software. The outputs of the Flex Set assays are the standard curves for each assay and the tabulated results for each analytes. Fig. 6 shows the bead positions of a 9-plex cell signaling Flex Set combination. Fig. 7 demonstrates the kinetics of T cell activation with CD3/CD28 treatment using Jurkat cells.
1. The BD CBA is not recommended for use on stream-in-air instruments where signal intensities may be reduced, adversely affecting assay sensitivity. Stream-in-air instruments include the FACStar Plus and FACSVantage flow cytometers.
2. The antibody-conjugated beads will settle out of suspension over time. It is necessary to vortex the vial vigorously for 3-5 s before taking a bead suspension aliquot.
3. The human Th1/Th2 cytokine standards vials are stable until the kit expiration date. Following reconstitution, store the freshly reconstituted 10X bulk standard at 2-8°C and use within 12 h.
4. When running experiments with higher order multiplexes use the following instructions for reconstituting the soluble protein Flex Set standards. For multiplex experiments involving 10-20 soluble protein Flex Set assays, reconstitute each standard vial with 0.1 mL of assay diluent to prepare a 20X bulk standard. For multiplex experiments involving more than 20 soluble protein Flex Set assays, pour each standard protein sphere into a 15-mL conical tube and reconstitute all spheres together in 2 mL of assay diluent to prepare a top standard mixture.
5. To calibrate the flow cytometer and quantitate test samples, it is necessary to run the cytokine standards and the cytometer setup controls in each experiment.
6. For Flex Set assays that will be acquired on a FACSCalibur flow cytometer, it is recommended that additional dilutions of the standard be prepared (i.e., 1:512 and 1:1024) as it is possible that in multiplex experiments containing a large number of assays, the top standard and 1:2 standard dilution will not be analyzable by the FCAP Array software. In those cases, the top standard and 1:2 standard dilutions can be run on the experiment but will be excluded from the final analysis in the FCAP Array software.
7. Cell lysates may be stored at -70°C for up to 6 mo. Multiple freeze/thaw treatments of sample should be avoided.
8. It is necessary to analyze CBA samples on the day of the experiment. Prolonged storage of samples, once the assay is complete, can lead to increased background and reduced sensitivity.
9. The phospho-specific cell signaling Flex Set assays cannot be used in the same assay well with the total protein cell signaling Flex Set assays. An updated assay compatibility chart for the cell signaling Flex Sets is available at www.bdbio-sciences.com/flexset.
1 Carson, R. T. and Vignali, D. A. (1999) Simultaneous quantitation of fifteen cytokines using a multiplexed flow cytometric assay. J. Immunol. Meths. 227,41-45.
2 Morgan, E., Varro, R., Sepulveda, H., et al. (2004) Cytometric bead array: a multiplexed assay platform with applications in various areas of biology. Clin. Immunol. 110, 252-266.
3 Stall, A., Sun, Q., Varro, R., et al. (1998) A single tube flow cytometric multibead assay for isotyping mouse monoclonal antibodies. Abstract 1877, Experimental Biology Meeting.
4 Lund-Johansen, F., Davis, K., Bishop, J. E., and Malefyt, R. de W. (2000) Flow cytometric analysis of immunoprecipitates: high-throughput analysis of protein phosphorylation and protein-protein interactions. Cytometry 39, 250-259.
5 Fulwyler, M. J., McHugh, T. M., Schwadron, R., et al. (1988) Immunoreactive bead (IRB) assay for the quantitative and simultaneous flow cytometric detection of multiple soluble analytes. Cytometry 2, 19.
6 McHugh, T. M. (1994) Flow microsphere immunoassay for the quantitative and simultaneous detection of multiple soluble analytes. Methods Cell Biol. 42, 575-595.
7. Camilla, C., Defoort, J. P., Delaage, M., et al. (1998) A new flow cytometry-based multi-assay system. 1. Application to cytokine immunoassays. Cytometry 8, 132.
8. Chen, R., Lowe, L., Wilson, J. D., et al. (1999) Simultaneous quantification of six human cytokines in a single sample using microparticle-based flow cytometric technology. Clin. Chem. 9, 1693-1694.
9 Fulton, R. J., McDade, R. L., Smith, P. L., Kienker, L. J., and Kettman, J. R. (1997) Advanced multiplexed analysis with the FlowMetrix system. Clin. Chem. 43, 1749-1756.
10 McHugh, T. M., Viele, M. K., Chase, E. S., and Recktenwald, D. J. (1997) The sensitive detection and quantitation of antibody to HCV using a microsphere-based immunoassay and flow cytometry. Cytometry 29, 106-112.
11 Faucher, S., Martel, A., Sherring, A., et al. (2004) Protein bead array for the detection of HIV-1 antibodies from fresh plasma and dried-blood-spot specimens. Clin. Chem. 50, 1250-1253.
12 Khan, I. H., Kendall, L. V., Ziman, M., et al. (2005) Simultaneous serodetection of 10 highly prevalent mouse infectious pathogens in a single reaction by multiplex analysis. Clin. Diagn. Lab. Immunol. 12, 513-519.
13 Cook, E. B., Stahl, J. L., Lowe, L., et al. (2001) Simultaneous measurement of six cytokines in a single sample of human tears using microparticle-based flow cytom-etry: allergics vs. non-allergics. J. Immunol. Meths. 254, 109-118.
14 Sonoda, S., Uchino, E., Nakao, K., and Sakamoto, T. (2006) Inflammatory cytokine of basal and reflex tears analysed by multicytokine assay. British J. Ophthalmology 90, 120-122.
15 Tarnok, A., Hambsch, J., Chen, R., and Varro, R. (2003) Cytometric bead array to measure six cytokines in twenty-five microliters of serum Clin. Chem. 49, 1000-1002.
16 Hodge, G., Hodge, S., Haslam, R., et al. (2004) Rapid simultaneous measurement of multiple cytokines using 100 microliter sample volumes—association with neonatal sepsis. Clin. Exp. Immunol. 137, 402-407.
17 Jatta, K., Wagsater,D., Norgren, L., Stenberg, B., and Sirsjo, A. (2005) Lipopoly-saccharide-induced cytokine and chemokine expression in human carotid lesions. J. Vasc. Res. 42, 266-271.
18 Penis, J. S., Yu, W. C., Leung, C. W., et al. (2004) Re-emergence of fatal human influenza A subtype H5N1 disease. Lancet 363, 617-619.
19 Rodriguez-Caballero, A., Garcia-Montero, A. C., Bueno, C., et al. (2004) A new simple whole blood flow cytometry based method for simultaneous identification of activated cells and quantitative evaluation of cytokines released during activation. Laboratory Investigation 84, 1387-1398.
20 Chen, L., Apgar, J., Huynh, L., et al. (2005) ZAP-70 directly enhances IgM signaling in chronic lymphocytic leukemia. Blood 105, 2036-2041.
Was this article helpful?